# What is the purpose of a wet leg?

What is the purpose of a wet leg?

## What is the purpose of a wet leg?

Wet leg method is used where vapor has a tendency to make the condensate and temperature of process is high or low from the atmospheric .

## What is a wet reference leg?

– A reference leg is used to compensate for vapor pressure in closed vessels. – A dry reference leg is called a Dry leg. – A Wet Leg is a reference leg filled with a liquid different from that in the vessel.

What is the difference between dry leg and wet leg?

Wet legs are used on differential pressure measurement where both sides are filled with liquid. Exception is level measurement in a pressurised vessel (boilers). Dry legs are used where the low side is exposed to atmosphere (using a DP transmitter for level measurement in a open tank.

### How do you calculate closed tank level?

The level in a closed vessel is calculated by the following formula: h = (p2 – p1)/ (ρ * g) p2 = hydrostatic pressure [bar] at depth h p1= pressure of the enclosed gas in the vessel [bar] ρ = density of the liquid [kg/m³] g = gravitational force or gravitational acceleration [m/s²] h = height of the liquid column [m] A …

### What is the advantage of using a wet reference leg over a dry reference leg?

Wet legs are used when the vapour in the tank will condense into liquid form. Likewise, dry legs are used when the vapour does not condense. While rela- tively simple in concept, impulse piping installations can be difficult to maintain.

What is URV and LRV?

range-value, lower (LRV) : The lowest value of the measured variable that a device is adjusted to measure. range-value, upper (URV) : The highest value of the measured variable that a device is adjusted to measure.

#### What is closed tank?

CPX closed tanks for waste water, drinking water or irrigation are available in different sizes from 1250 to 6000 litres. The tanks are suitable for both below-ground and above-ground installation. Our green, child-proof hatch blends into its surroundings and comes with all tanks as standard.

How do you calculate pressure in a closed tank?

In the case of the water stored in a tank, the pressure at its bottom is the weight acting on a unit area of the surface where the tank is kept. To translate that into an equation: Pressure = weight/area, and weight = mass (m) * acceleration due to gravity (g). This means pressure = m * g/ area.

## What is the range of DP?

The calibration range may be the same or differ from the instrument range. For example, a DP transmitter may have a nameplate instrument range of 0–700 psig and output of 4 – 20 mA. However, if an engineer has determined the instrument will be calibrated for 0-to-300 psig = 4-to-20 mA.

## What is span limit of pressure transmitter?

Span – The difference between the lowest point and the highest point in a configured measurement range. (For example, if a 250 psi transmitter is adjusted to measure a range of 50 psi to 200 psi, then the span is 150 psi.)